These specifications are for the Pulsar Nova Model 43 and Nova Model 44 Integrating Sound Level Meter.
Most occupational noise regulations demand the use of a Type 2 Sound Level Meter. Both the Model 43 and Model 44 meet or exceed these requirements. Note that you should also have a suitable Sound Level Calibrator.
If you have any questions about the standards your meter needs to meet then please call on 888-206-4377 or email on email@example.com.
20 to 140 dB RMS in a Single Range (no need for range selection)
Model 43 Type 1 noise floor < 18 dB(A)
Model 44 Type 2 noise floor < 21 dB(A)
"F" Fast, "S" Slow and "I" Impulse, all measured simultaneously.
"A", "C" and "Z" weighting, all measured simultaneously. (Z weighting is flat from 20Hz to 20kHz ±1.5dB)
Noise Measurement Parameters
LXY LXYmax and LXYmin
LXeq LXpeak LXE LAIeq
where X = frequency weighting (A, C or Z) and Y = time weighting (Fast, Slow or Impulse)
6 x AA for typically 30 hours operation with Alkaline and 50 hours with Lithium
5V to 15V via Multipin I/O socket
Weight 440g excluding batteries
Size 280mm x 80mm x 42mm with preamplifier and microphone fitted
USB 2.0 Mini-B
AC output via 3.5mm stereo jack
AC and DC output vi Multipin I/O socket
Standard 1/4" threaded socket
Diecast aluminum with silicon elastomer keypad and soft touch back
High resolution anti-glare OLED
Ambient light sensor for automatic brightness adjustment
Illuminated keypad for low-light operation
Operating Temperature: -10°C to +50°C
Storage Temperature: -20°C to +60°C
Humidity: Up to 95% RH Non-condensing
IEC 61672-1:2002 and IEC 61672-2:2003
Except where modified by EN 61000-6-1:2007
Data Logging and Storage (options)
Removable memory size is 4GB
English, French, German, Italian and Spanish.